|Miniaturized Laser Diode Solution for the Medical Market|
|ㆍ작성일: 2012-01-10 (화) 01:46||ㆍ조회: 218|
Developing a Miniaturized Laser Diode
Solution for the Medical Market
Posted in Features, Surgical Lasers, Treatment Devices on Tuesday, November 01 2011
According to Pantec Biosolutions AG (Liechtenstein, Europe), the global aesthetic market, which includes skin rejuvenation, is expected to grow from a $4.4 billion market in 2010 to a $7.5 billion market in 2015. Meanwhile, the market for transdermal drug delivery is growing rapidly and is expected to be a multi-billion dollar market by 2015. Designed for medical professionals and consumers, these new devices offer a precise delivery method for a variety of medical applications, such as skin rejuvenation, in-vitro fertilization, and vaccinations.
The following article discusses how Oclaro (San Jose, CA) worked with Pantec Biosolutions AG to develop this customized quasi-continuous wave (QCW) sub-assembly laser diode solution.
Market and Application Can Drive End-Product Requirements
An industrial application usually requires higher power and higher brightness, as well as a laser diode lifetime of several ten-thousand hours. Medical devices, on the other hand, tend to have lower requirements on diode lifetime and are more sensitive on parameters such as cost, efficiency, thermal management, and size. In this example, a small handheld batterypowered laser device needed to be developed. Oclaro was able to leverage its full skill set from epitaxial design of semiconductor lasers, wafer processing, optical coating and facet passivation, as well as assembly design. The Pantec Biosolutions’ team was responsible for designing the solid state laser, as well as the overall application know-how, including clinical studies and compliance with laser safety and medical regulations.
This particular project with Pantec Biosolutions focused on its laser-based transdermal delivery platform P.L.E.A.S.E.®(Precise Laser Epidermal System). The first device using this new platform is the newly-released P.L.E.A.S.E.® Professional (Fig 1), a tabletop medical laser device targeted mainly for the dermatologic and aesthetic markets. It creates precisely controlled aqueous micropores through the stratum corneum into the epidermis (Fig. 2). An intelligent graphical user interface, together with the CE mark and the laser class 1 grading, make it simple and safe for the medical professional to use. At a later time, this device will be complemented by the P.L.E.A.S.E.®Private (Fig 4), a battery-powered handheld medical laser device targeted mainly for drug delivery.
Meeting Small Packaging Requirements is Crucial
For the epitaxial design of the device, the focus was on optimizing power conversion efficiency and thermal robustness. The resulting structure is characterized by a single InGaAs quantum well, and an AlGaAs wave guide. Oclaro’s E2 facet passivation technology, which protects the front facet of the bar against Catastrophic Optical Damage, enables reliable operation of multimode devices at high ex-facet power densities of 120mW/μm or more [2,3]. The passivated mirror facets are coated with high anti-reflection coatings for optimized device performance.
For qcw bars operating at 500A drive current and about 600W output power, 200-μs pulse width and 5-8% duty cycle was performed with bars of the present 970nm chip configuration. No significant degradation was observed after 370 million pulses. The most critical assembly steps were carried out on fully automated die bonding systems, which ensures a high alignment precision and reproduceable solder interfaces with good thermal properties and a homogeneous stress distribution. The thermal properties were simulated using the finite-elementmethod. Telecom-grade AuSn (gold tin) hard solder makes the product suitable for the most demanding hard-pulse operation mode. The laser diode sub-assembly is shown in Fig. 3.
A key requirement for the design of a monolithic solid state laser without any active adjustment possibility is to ensure stable operation within a wide range of heat load generated in the laser cavity.
Developing a Miniaturized Diode-Pumped Laser System
The 1-mm-diameter Er:YAG laser rod is side pumped by a single 10-mm-wide 970nm laser diode bar mounted on a 10×12×5 mm3 conductively cooled platform. The target application requires reliable laser diode operation at 300W, 200μs, a duty cycle of 10%, and a base plate temperature of 20–30 °C. For efficient pumping of the rod, a laser diode without wavelength stabilization was required to be designed for high thermal robustness. The high power of the laser diode and the small laser rod diameter enable side pumping of Er:YAG despite its low emission cross-section. In order to operate the laser efficiently, the losses in the cavity need to be minimized. This has been achieved by using a monolithic laser cavity. As a further advantage, the monolithic cavity requires no alignment, and therequired robustness against mechanical shocks is easily obtained without the need for time-consuming adjustments.
Present and Future Target Applications
According to Pantec, the target applications for the medical diode-pumped solid state laser system are transdermal delivery of pharmaceutical and cosmeceutical products and Laser Dynamic Therapy® (LDT). Microporation offers advantages in the delivery of topically applied systemic treatments such as large molecule protein drugs to treat precancerous skin lesions that at present cannot be adequately delivered in topical application. In addition, in vitro analyses have shown that P.L.E.A.S.E. microporation can facilitate the rapid transdermal delivery of various topical anticancer drugs at adequate levels for the efficient treatment of pre-cancerous lesions, suggesting accelerated biological response and increased efficiency.
It is also appropriate for combination treatments where transdermal delivery of cosmeceuticals may be appropriate for certain dermatological conditions. This approach is less invasive than conventional alternatives, and offers faster healing and a shorter downtime with the potential for improved clinical outcomes.
Pantec Biosolutions expects that the accelerated biological response within the target structure will result in a faster clearance of the treated tumor lesions, compared to known conventional topical treatments. The ultimate goal is to improve patient convenience and compliance by giving dermatologists several options to increase treatment efficiency of pre-cancerous lesions and to reduce treatment time and pain associated with standard photodynamic therapy (PDT) or conventional ointment therapies. The system can either enhance the therapeutic outcome of standard PDT by increased permeation of photosensitizers deeper into the dermis, or by completely replacing the currently used methods with Laser Dynamic Therapy.
In conclusion, the successful development of an application-specific product requires close cooperation on all levels with the tight interaction of R&D teams. The semiconductor as well as laser diode assembly have to be matche d to the requirements of the end product, which can vary strongly depending on market and application.
This article was written by Christian Naumer, Senior Product Line Manager for Oclaro, Inc., San Jose, CA. For more information, visit http://info.hotims.com/34460-164.
 M. Krejci, A. Heinrich, J. Müller, T. Bragagna and N. Lichtenstein, "Miniaturized high power Er:YAG solid state laser pumped by a single laser diode bar", Proc. SPIE 7912, 79121D (2011); doi:10.1117/12.873576
 D. Jaeggi, C. Naumer, “Diversity is strength for high-power laser diodes”,optics.org/OLE, 1/2009
 C. Naumer, N. Lichtenstein, B. Schmidt, P. Bruns, F. Kubacki, P. Harten, O. Lissotschenko,“Brightness Required!”,Laser & Photonik, 3/2007
 C.Naumer, D.Jaeggi, N.Lichtenstein, B.Schmidt, “Setting the Pace in Industrial Laser Systems”, Laser & Photonik, 4/2006
 N. H. Zech M.D., M. Murtinger M.D., P. Uher M.D., “Pregnancy after ovarian superovulation by transdermal delivery of follicle-stimulating hormone”, Fertility and Sterility, Volume 95, Issue 8, 30 June 2011, Pages 2784-2785
 Y.G. Bachhav, S. Summer, A. Heinrich, T. Bragagna, C. Böhler, Y.N. Kalia, „Effect of controlled laser microporation on drug transport kinetics into and across the skin”, Journal of Controlled Release, Volume 146, Issue 1, 17 August 2010, Pages 31-36
 J. Yu, Y.G. Bachhav, S. Summer, A. Heinrich, T. Bragagna, C. Böhler, Y.N. Kalia, “Using controlled laser-micro -poration to increase transdermal delivery of prednisone”, Journal of Controlled Release, Volume 148, Issue 1, 20 November 2010, Pages e71-e73